LW-10 wavelength meter key performances
Here below are presented the measurements performed with our LW-10 wavelength meter on three typical parameters: resolution, stability over time and temperature stability.
20 MHz resolution
20 MHz peak-to-peak Resonator Triangular ramps measured at 780 nm with SpectraResolver software on a Ti:Sa laser.
Stability over time
Measurement of 800 nm stabilized wavelength (Ti:Sa laser locked with 0.0001 nm tolerance) and temperature measurement in thermal chamber (23.5°C +/- 0.1°C stabilized temperature).
Measurement of 780 nm stabilized wavelength (Ti:Sa laser locked with 0.0001 nm tolerance) and temperature measurement in the thermal chamber: stabilized temperature 23.0°C +/- 0.1°C during 20 hours, 16°C to 30°C with +/- 2°C per hour temperature ramps, and 15 hours in stabilized temperature.